|K35200 Humidity Cabinet for Testing Rust Protection|
Tests the ability of metal preservatives to prevent steel panels from rusting under conditions of high humidity. Polished steel panels are immersed in the sample oil and then suspended in the humidity cabinet for a specified test period.
Features and Benefits
Conforms to ASTM D1748 and FTM 791-5310 specifications
Click on the image to view a larger photo of the instrument.
|K35200 Humidity Cabinet for Testing Rust Protection
Produces a moisture saturated atmosphere with continuous condensation at a constant 120°F (48.9°C) for 33 steel test specimens. Test panels are suspended on a 1/3rpm rotating stage. Air flow and water level control systems maintain required conditions inside the cabinet per Mil. Spec. and ASTM specifications. Air temperature is maintained at 120 ±2°F (48.9 ±1.1°C) by a digital LCD electronic controller. A continuous heater circuit assists the control heater in bringing the cabinet up to temperature prior to testing. Overtemperature protection is provided by an adjustable digital thermostat which cuts off power to the cabinet in case of overheating.
Cabinet interior is stainless steel lined and all interior components are of stainless steel or chrome plated steel construction. Hinged cover consists of two layers of desized cotton cloth mounted on a metal frame. Oil and condensate dripping from the specimens are collected in a drip pan and piped to an external drain.
Conforms to the specifications of:
ASTM D1748; FTM 791-5310
Capacity: 33 rust test specimens
Water Level Control: 8 in. (203mm)
Temperature Control Stability: ±2°F (± 1.1°C) (air temperature)
Heater Range: 0-1500W
Air Metering: 0.878±0.02832m3/h at standard temperature and pressure (31±1 ft3/h)
Air Distribution: 20-diffuser manifold
Rotating Stage: 1/3rpm
115V 60Hz, Single Phase, 13.0A
220-240V 50Hz or 60Hz, Single Phase, 6.8A
Monel Test Specimen Hooks (33 sets)
Net Weight: 206 lbs (93.4kg)
Shipping Weight: 279 lbs (126.6kg)
Dimensions: 41 Cu. ft.
|When inquiring about NIST traceable certified thermometers, please refer to the catalog number for the corresponding thermometer and replace the middle three zeros in the catalog number with 004. Example: 251-000-01C ASTM 1C Thermometer would be 251-004-01C ASTM 1C Certified Thermometer.|